You Asked for Simplicity. We Delivered.
- SureBP technology can record blood pressure in just 15 seconds
- BP averaging helps ensure accurate hypertension detection
- SureTemp Plus oral/axillary thermometry
- Option for Nonin® SpO2 technology
- 7" color touchscreen for a simplified workflow
- Options for documenting patient observations (like respiration rate and BMI)
- Ships with one adult blood pressure cuff
Does Movement Matter for BP Readings?
A patient's own movement—voluntary or accidental—can make blood pressure readings inaccurate. That’s why it’s important to equip your practice with solutions that can capture accurate blood pressure readings in the presence of movement. Many Hillrom vital signs solutions use the Welch Allyn SureBP® algorithm, which can do just that, helping to improve your patient data quality and increase your clinical efficiency.
You asked for simplicity. We delivered.
Education & Documentation
Get in the know to get the most value out of your solution.
Learn how to perform an exam with the Welch Allyn Spot Vital Signs 4400 Device.
Discover how the Welch Allyn Spot Vital Signs 4400 Device's blood pressure averaging function can help you accurately detect hypertension.
Find out how our Welch Allyn Spot Vital Signs 4400 Device can improve patient data collection by connecting to your EMR.
This video discusses how to set up your Welch Allyn Spot Vital Signs 4400 Device and its accessories.
Review troubleshooting and basic service steps for the Welch Allyn Spot Vital Signs 4400 Device in this helpful video.
You'll find tips for navigating and setting up your Welch Allyn Spot Vital Signs 4400 Device in this easy-to-follow video.
Does Movement Matter? Exploring the Effect of Movement on Blood Pressure Readings with the Welch Allyn® SureBP® Algorithm
Vital Considerations: Automatic Blood Pressure Devices Can Deliver Long-Term Benefits in Primary Care
To Average or Not To Average? Help Improve Hypertension Detection with Blood Pressure Averaging
What cleaning agents can I use on my Spot Vital Signs 4400 Device's touchscreen?keyboard_arrow_down
You can use wipes on the touchscreen monitor surface, as long as the bleach concentration does not exceed 10%.
Note: When cleaning the device, avoid using cloths or solutions that include quaternary ammonium compounds (ammonium chlorides) or glutaraldehyde-based disinfectants.
The following agents are compatible with the monitor. Follow the cleaning agent manufacturer’s guidelines:
- Accel INTERVention
- Accel TB
- Clinell® Universal Wipes
- Clorox® HealthCare Bleach Germicidal Cleaner
- Oxivir® TB
- Sani-Cloth® Plus
- Sani-Cloth® Bleach
- Super HDQ® L10, dilution rate of ½ oz. per gallon of water (1:256), applied to a clean cloth
- Super Sani-Cloth®
- Tuffie 5 Cleaning Wipes
- Virex® II (256), dilution rate of ½ oz. per gallon of water (1:256), applied to a clean cloth
- 10% bleach solution, 0.5% - 1% sodium hypochlorite, applied to a clean cloth
- 70% isopropyl alcohol solution
Note: Disinfect according to your facility’s protocols and standards, or local regulations.
How do I create my own Blood Pressure Averaging programs on my Spot Vital Signs® 4400 Device?keyboard_arrow_down
The device comes pre-loaded with a standard Blood Pressure Averaging program. Should you wish to change this, access it in the Advanced Settings menu, under Parameters and Averaging. You can also view this video to learn about setting up Blood Pressure Averaging programs.
How do I access the Advanced Settings menu on my Spot Vital Signs® 4400 Device?keyboard_arrow_down
Access the Advanced Settings menu from the Navigation menu, under Settings and Advanced Settings. No code is required for this menu option; simply press the button to enter.
Where can I get extra blood pressure cuffs for my device?keyboard_arrow_down
The Spot Vital Signs 4400 Device uses our Welch Allyn® FlexiPort® BP cuff system, so it's compatible with BP cuffs for any other Welch Allyn monitoring devices that you might have in your facility. Contact your Hillrom representative for details on how to order additional cuffs.